Sensory substitution is a change of the characteristics of one sensory modality into stimuli of another sensory modality.
A sensory substitution system consists of three parts: a sensor, a coupling system, and a stimulator. The sensor records stimuli and gives them to a coupling system which interprets these signals and transmits them to a stimulator. In case the sensor obtains signals of a kind not originally available to the bearer it is a case of sensory augmentation. Sensory substitution concerns human perception and the plasticity of the human brain; and therefore, allows us to study these aspects of neuroscience more through neuroimaging.
It is hoped[according to whom?] that sensory substitution systems can help people by restoring their ability to perceive certain defective sensory modality by using sensory information from a functioning sensory modality.
The idea of sensory substitution was introduced in the '80s by Paul Bach-y-Rita as a means of using one sensory modality, mainly taction, to gain environmental information to be used by another sensory modality, mainly vision. Thereafter, the entire field was discussed by Chaim-Meyer Scheff in "Experimental model for the study of changes in the organization of human sensory information processing through the design and testing of non-invasive prosthetic devices for sensory impaired people". The first sensory substitution system was developed by Bach-y-Rita et al. as a means of brain plasticity in congenitally blind individuals. After this historic invention, sensory substitution has been the basis of many studies investigating perceptive and cognitive neuroscience. Since then, sensory substitution has contributed to the study of brain function, human cognition and rehabilitation.
When a person becomes blind or deaf they generally do not lose the ability to hear or see; they simply lose their ability to transmit the sensory signals from the periphery (retina for visions and cochlea for hearing) to brain. Since the vision processing pathways are still intact, a person who has lost the ability to retrieve data from the retina can still see subjective images by using data gathered from other sensory modalities such as touch or audition.
In a regular visual system, the data collected by the retina is converted into an electrical stimulus in the optic nerve and relayed to the brain, which re-creates the image and perceives it. Because it is the brain that is responsible for the final perception, sensory substitution is possible. During sensory substitution an intact sensory modality relays information to the visual perception areas of the brain so that the person can perceive sight. With sensory substitution, information gained from one sensory modality can reach brain structures physiologically related to other sensory modalities. Touch-to-visual sensory substitution transfers information from touch receptors to the visual cortex for interpretation and perception. For example, through fMRI, one can determine which parts of the brain are activated during sensory perception. In blind persons, it is seen that while they are only receiving tactile information, their visual cortex is also activated as they perceive sight objects. Touch-to-touch sensory substitution is also possible, wherein information from touch receptors of one region of the body can be used to perceive touch in another region. For example, in one experiment by Bach-y-Rita,touch perception was able to be restored in a patient who lost peripheral sensation due to leprosy.
In order to achieve sensory substitution and stimulate the brain without intact sensory organs to relay the information, machines can be used to do the signal transduction, rather than the sensory organs. This brain–machine interface collects external signals and transforms them into electrical signals for the brain to interpret. Generally, a camera or a microphone is used to collect visual or auditory stimuli that are used to replace lost sight and hearing, respectively. The visual or auditory data collected from the sensors is transformed into tactile stimuli that are then relayed to the brain for visual and auditory perception. This and all types of sensory substitution are only possible due to neuroplasticity.
Brain plasticity refers to the brain's ability to adapt to a changing environment, for instance to the absence or deterioration of a sense. It is conceivable that cortical remapping or reorganization in response to the loss of one sense may be an evolutionary mechanism that allows people to adapt and compensate by using other senses better. Functional imaging of congenitally blind patients showed a cross-modal recruitment of the occipital cortex during perceptual tasks such as Braille reading, tactile perception, tactual object recognition, sound localization, and sound discrimination. This may suggest that blind people can use their occipital lobe, generally used for vision, to perceive objects through the use of other sensory modalities. This cross modal plasticity may explain the often described tendency of blind people to show enhanced ability in the other senses.
Perception versus sensingEdit
While considering the physiological aspects of sensory substitution, it is essential to distinguish between sensing and perceiving. The general question posed by this differentiation is: Are blind people seeing or perceiving to see by putting together different sensory data? While sensation comes in one modality – visual, auditory, tactile etc. – perception due to sensory substitution is not one modality but a result of cross-modal interactions. It is therefore concluded that while sensory substitution for vision induces visual-like perception in sighted individuals, it induces auditory or tactile perception in blind individuals. In short, blind people perceive to see through touch and audition with sensory substitution.
Applications are not restricted to handicapped persons, but also include artistic presentations, games, and augmented reality. Some examples are substitution of visual stimuli to audio or tactile, and of audio stimuli to tactile. Some of the most popular are probably Paul Bach-y-Rita's Tactile Vision Sensory Substitution (TVSS), developed with Carter Collins at Smith-Kettlewell Institute and Peter Meijer's Seeing with Sound approach (The vOICe). Technical developments, such as miniaturization and electrical stimulation help the advance of sensory substitution devices.
In sensory substitution systems, we generally have sensors that collect the data from the external environment. This data is then relayed to a coupling system that interprets and transduces the information and then replays it to a stimulator. This stimulator ultimately stimulates a functioning sensory modality. After training, people learn to use the information gained from this stimulation to experience a perception of the sensation they lack instead of the actually stimulated sensation. For example, a leprosy patient, whose perception of peripheral touch was restored, was equipped with a glove containing artificial contact sensors coupled to skin sensory receptors on the forehead (which was stimulated). After training and acclimation, the patient was able to experience data from the glove as if it was originating in the fingertips while ignoring the sensations in the forehead.
To understand tactile sensory substitution it is essential to understand some basic physiology of the tactile receptors of the skin. There are five basic types of tactile receptors: Pacinian corpuscle, Meissner's corpuscle, Ruffini endings, Merkel nerve endings, and free nerve endings. These receptors are mainly characterized by which type of stimuli best activates them, and by their rate of adaptation to sustained stimuli. Because of the rapid adaptation of some of these receptors to sustained stimuli, those receptors require rapidly changing tactile stimulation systems in order to be optimally activated. Among all these mechanoreceptors Pacinian corpuscle offers the highest sensitivity to high frequency vibration starting from few 10s of Hz to a few kHz with the help of its specialized mechanotransduction mechanism.
There have been two different types of stimulators: electrotactile or vibrotactile. Electrotactile stimulators use direct electrical stimulation of the nerve ending in the skin to initiate the action potentials; the sensation triggered, burn, itch, pain, pressure etc. depends on the stimulating voltage. Vibrotactile stimulators use pressure and the properties of the mechanoreceptors of the skin to initiate action potentials. There are advantages and disadvantages for both these stimulation systems. With the electrotactile stimulating systems a lot of factors affect the sensation triggered: stimulating voltage, current, waveform, electrode size, material, contact force, skin location, thickness and hydration. Electrotactile stimulation may involve the direct stimulation of the nerves (percutaneous), or through the skin (transcutaneous). Percutaneous application causes additional distress to the patient, and is a major disadvantage of this approach. Furthermore, stimulation of the skin without insertion leads to the need for high voltage stimulation because of the high impedance of the dry skin, unless the tongue is used as a receptor, which requires only about 3% as much voltage. This latter technique is undergoing clinical trials for various applications, and been approved for assistance to the blind in the UK. Alternatively, the roof of the mouth has been proposed as another area where low currents can be felt.
Electrostatic arrays are explored as human-computer interaction devices for touch screens. These are based on a phenomenon called electrovibration, which allows microamperre-level currents to be felt as roughness on a surface.
Vibrotactile systems use the properties of mechanoreceptors in the skin so they have fewer parameters that need to be monitored as compared to electrotactile stimulation. However, vibrotactile stimulation systems need to account for the rapid adaptation of the tactile sense.
Another important aspect of tactile sensory substitution systems is the location of the tactile stimulation. Tactile receptors are abundant on the fingertips, face, and tongue while sparse on the back, legs and arms. It is essential to take into account the spatial resolution of the receptor as it has a major effect on the resolution of the sensory substitution. A high resolution pin-arrayed display is able to present spatial information via tactile symbols, such as city maps and obstacle maps.
Below you can find some descriptions of current tactile substitution systems.
One of the earliest and most well known form of sensory substitution devices was Paul Bach-y-Rita's TVSS that converted the image from a video camera into a tactile image and coupled it to the tactile receptors on the back of his blind subject. Recently, several new systems have been developed that interface the tactile image to tactile receptors on different areas of the body such as the on the chest, brow, fingertip, abdomen, and forehead. The tactile image is produced by hundreds of activators placed on the person. The activators are solenoids of one millimeter diameter. In experiments, blind (or blindfolded) subjects equipped with the TVSS can learn to detect shapes and to orient themselves. In the case of simple geometric shapes, it took around 50 trials to achieve 100 percent correct recognition. To identify objects in different orientations requires several hours of learning.
A system using the tongue as the human-machine interface is most practical. The tongue-machine interface is both protected by the closed mouth and the saliva in the mouth provides a good electrolytic environment that ensures good electrode contact. Results from a study by Bach-y-Rita et al. show that electrotactile stimulation of the tongue required 3% of the voltage required to stimulate the finger. Also, since it is more practical to wear an orthodontic retainer holding the stimulation system than an apparatus strapped to other parts of the body, the tongue-machine interface is more popular among TVSS systems.
This tongue TVSS system works by delivering electrotactile stimuli to the dorsum of the tongue via a flexible electrode array placed in the mouth. This electrode array is connected to a Tongue Display Unit [TDU] via a ribbon cable passing out of the mouth. A video camera records a picture, transfers it to the TDU for conversion into a tactile image. The tactile image is then projected onto the tongue via the ribbon cable where the tongue's receptors pick up the signal. After training, subjects are able to associate certain types of stimuli to certain types of visual images. In this way, tactile sensation can be used for visual perception.
Sensory substitutions have also been successful with the emergence of wearable haptic actuators like vibrotactile motors, solenoids, peltier diodes, etc. At the Center for Cognitive Ubiquitous Computing at Arizona State University, researchers have developed technologies that enable people who are blind to perceive social situational information using wearable vibrotactile belts (Haptic Belt) and gloves (VibroGlove). Both technologies use miniature cameras that are mounted on a pair of glasses worn by the user who is blind. The Haptic Belt provides vibrations that convey the direction and distance at which a person is standing in front of a user, while the VibroGlove uses spatio-temporal mapping of vibration patterns to convey facial expressions of the interaction partner. Alternatively, it has been shown that even very simple cues indicating the presence or absence of obstacles (through small vibration modules located at strategic places in the body) can be useful for navigation, gait stabilization and reduced anxiety when evolving in an unknown space. This approach, called the "Haptic Radar" has been studied since 2005 by researchers at the University of Tokyo in collaboration with the University of Rio de Janeiro.
Neuroscientist David Eagleman presented a new device for sound-to-touch hearing at TED in 2015; his laboratory research then expanded into a company based in Palo Alto, California, called NeoSensory. NeoSensory devices capture sound and turn them into high-dimensional patterns of touch on the skin.
Experiments by Schurmann et al. show that tactile senses can activate the human auditory cortex. Currently vibrotactile stimuli can be used to facilitate hearing in normal and hearing-impaired people. To test for the auditory areas activated by touch, Schurmann et al. tested subjects while stimulating their fingers and palms with vibration bursts and their fingertips with tactile pressure. They found that tactile stimulation of the fingers lead to activation of the auditory belt area, which suggests that there is a relationship between audition and tactition. Therefore, future research can be done to investigate the likelihood of a tactile–auditory sensory substitution system. One promising invention is the 'Sense organs synthesizer' which aims at delivering a normal hearing range of nine octaves via 216 electrodes to sequential touch nerve zones, next to the spine.
Some people with balance disorders or adverse reactions to antibiotics suffer from bilateral vestibular damage (BVD). They experience difficulty maintaining posture, unstable gait, and oscillopsia. Tyler et al. studied the restitution of postural control through a tactile for vestibular sensory substitution. Because BVD patients cannot integrate visual and tactile cues, they have a lot of difficulty standing. Using a head-mounted accelerometer and a brain-machine interface that employs electrotactile stimulation on the tongue, information about head-body orientation was relayed to the patient so that a new source of data is available to orient themselves and maintain good posture.
Tactile–tactile to restore peripheral sensationEdit
Touch to touch sensory substitution is where information from touch receptors of one region can be used to perceive touch in another. For example, in one experiment by Bach-y-Rita, the touch perception was restored in a patient who lost peripheral sensation from leprosy. For example, this leprosy patient was equipped with a glove containing artificial contact sensors coupled to skin sensory receptors on the forehead (which was stimulated). After training and acclimation, the patient was able to experience data from the glove as if it was originating in the fingertips while ignoring the sensations in the forehead. After two days of training one of the leprosy subjects reported "the wonderful sensation of touching his wife, which he had been unable to experience for 20 years."
Tactile feedback system for prosthetic limbsEdit
The development of new technologies has now made it plausible to provide patients with prosthetic arms with tactile and kinesthetic sensibilities. While this is not purely a sensory substitution system, it uses the same principles to restore perception of senses. Some tactile feedback methods of restoring a perception of touch to amputees would be direct or micro stimulation of the tactile nerve afferents.
Other applications of sensory substitution systems can be seen in function robotic prostheses for patients with high level quadriplegia. These robotic arms have several mechanisms of slip detection, vibration and texture detection that they relay to the patient through feedback. After more research and development, the information from these arms can be used by patients to perceive that they are holding and manipulating objects while their robotic arm actually accomplishes the task.
Auditory sensory substitution systems like the tactile sensory substitution systems aim to use one sensory modality to compensate for the lack of another in order to gain a perception of one that is lacking. With auditory sensory substitution, visual or tactile sensors detect and store information about the external environment. This information is then transformed by brain-machine interfaces into auditory signals that are relayed via the auditory receptors to the brain.
Auditory vision substitutionEdit
Auditory vision substitution aims to use the sense of hearing to convey visual information to the blind.
The vOICe Auditory DisplayEdit
This section contains content that is written like an advertisement. (January 2018) (Learn how and when to remove this template message)
The vOICe Auditory Display Technology is one of several approaches towards sensory substitution (vision substitution) for the blind that aims to provide synthetic vision analogous to seeing their surroundings using sound waves by means of a non-invasive visual prosthesis.
The vOICe converts live camera views from a video camera into soundscapes, patterns of scores of different tones at different volumes and pitches emitted simultaneously. This system uses general video to audio mapping by associating height to pitch and brightness with loudness in a left-to-right scan of any video frame. Views are typically refreshed about once per second with a typical image resolution of up to 60 x 60 pixels as can be proven by spectrographic analysis.
The ultimate goal is to provide synthetic vision with truly visual sensations by exploiting the neural plasticity of the human brain. The system does not require any surgery. Its interface to the mind is simply headphones. Over time with practice, the processing within the brain is gradually sent down to the subconscious levels and becomes automatic.
Neuroscience research has shown that the visual cortex of even adult blind people can become responsive to sound, and "seeing with sound" might reinforce this in a visual sense with live video from a head-mounted camera encoded in sound. The extent to which cortical plasticity indeed allows for functionally relevant rewiring or remapping of the human brain is still largely unknown and is being investigated in an open collaboration with research partners around the world.
One suggestion for increasing the relative efficiency of the resulting visual stimuli is to adjust the visual field by using an accelerometer to provide a steady image even if the head is moved, which is implemented in its Android edition. Connecting an infrared sensor to adjust the camera position to match eye movements is an option for the Windows edition (though affordable mobile eye-trackers are not yet on the market).
The technology of the vOICe was invented in the 1990s by Peter Meijer. It has been positively and widely reviewed in media and also in various high-quality peer-reviewed scientific journals, far extensive than those partially referenced in the current section, and maintained by the inventor's website itself.
Unlike other models and prototypes aiming to do this particular task, this project is developed within a budget of 4 US dollars, and is aimed for the vast majority of partially or fully blind people living in developing and underdeveloped countries, who can't access equipment worth hundreds of dollars.
Project BATEYE uses an ultrasonic sensor mounted on to a wearable pair of glasses that measures the distance to the nearest object and relays it to an Arduino board. The Arduino board then processes the measurements and then plays a tone (150–15000 Hz) for the respective distance(2 cm to 4m) till the data from the second ultrasonic pulse (distance) comes in, and then the same process gets repeated. This cycle is repeated almost every 5 milliseconds. The person hears sound that changes according to the distance to the nearest object. The head provides a 195-degree swivel angle and the ultrasonic sensor detects anything within a 15-degree angle. Using systematic, cognitive and computational approach of neuroscience, with the hypothesis that the usage of the occipital lobe of blind people goes into processing other sensory feedback, and using the brain as a computational unit, the machine relies on the brain processing the tone produced every 14 mS to its corresponding distance and producing a soundscape corresponding to the tones and the body navigating using the same.During experimentation, the test subject could detect obstacles as far away as 2 – 3m, with horizontal or vertical movements of the head the blindfolded test subject could understand the basic shape of objects without touching them, and the basic nature of the obstacles.
Primarily, this research is open sourced and published (see: [Debargha Ganguly. (2016); DEVELOPING AN ECONOMIC SYSTEM THAT CAN GIVE A BLIND PERSON BASIC SPATIAL AWARENESS AND OBJECT IDENTIFICATION. Int. J. of Adv. Res. 4 (11). 2003–2008] (ISSN 2320-5407).
This is a part of the currently undertaken research by Debargha Ganguly, known as Project Basics – an effort to improve the basic standard of living of people of developing and underdeveloped countries, which has also included Project Awaaz – developing a low cost, non-movement restrictive plugin for hand movement to speech conversion.
The EyeMusic, represents high locations on the image as high-pitched musical notes on a pentatonic scale, and low vertical locations as low-pitched musical notes on a pentatonic scale. The user wears a miniature camera connected to a small computer (or smartphone) and stereo headphones. The images are converted into "soundscapes", using a predictable algorithm, allowing the user to listen to and then interpret the visual information coming from the camera.
The EyeMusic conveys color information by using different musical instruments for each of the following five colors: white, blue, red, green, yellow; Black is represented by silence. The EyeMusic currently employs an intermediate resolution of 30×50 pixels. An auditory cue is sounded at the beginning of each left-to-right scan of the image. (1) the higher musical notes represent pixels that are located higher on the y-axis of an image, (2) the timing of the sound after the cue indicates the x-axis location of the pixel (that is, an object located on the left of the image will be "sounded" earlier on than an object located further on the right), and (3) different colors are represented by different musical instruments. In addition, subjects who used the EyeMusic dynamically were able to accurately reach for an object perceived via the SSD.
This project, presented in 2015, proposes a new versatile mobile device and a sonification method specifically designed to the pedestrian locomotion of the visually impaired. It sonifies in real-time spatial information from a video stream acquired at a standard frame rate. The device is composed of a miniature camera integrated into a glasses frame which is connected to a battery-powered minicomputer worn around the neck with a strap. The audio signal is transmitted to the user via running headphones. This system has two operating modes. With the first mode, when the user is static, only the edges of the moving objects are sonified. With the second mode, when the user is moving, the edges of both static and moving objects are sonified. Thus, the video stream is simplified by extracting only the edges of objects that can become dangerous obstacles. The system enables the localization of moving objects, the estimation of trajectories, and the detection of approaching objects.
Another successful visual-to-auditory sensory substitution device is the Prosthesis Substituting Vision for Audition (PSVA). This system utilizes a head-mounted TV camera that allows real-time, online translation of visual patterns into sound. While the patient moves around, the device captures visual frames at a high frequency and generates the corresponding complex sounds that allow recognition. Visual stimuli are transduced into auditory stimuli with the use of a system that uses pixel to frequency relationship and couples a rough model of the human retina with an inverse model of the cochlea.
The sound produced by this software is a mixture of sinusoidal sounds produced by virtual "sources", corresponding each to a "receptive field" in the image. Each receptive field is a set of localized pixels. The sound's amplitude is determined by the mean luminosity of the pixels of the corresponding receptive field. The frequency and the inter-aural disparity are determined by the center of gravity of the co-ordinates of the receptive field's pixels in the image (see "There is something out there: distal attribution in sensory substitution, twenty years later"; Auvray M., Hanneton S., Lenay C., O'Regan K. Journal of Integrative Neuroscience 4 (2005) 505-21). The Vibe is an Open Source project hosted by Sourceforge.
Other approaches to the substitution of hearing for vision use binaural directional cues, much as natural human echolocation does. An example of the latter approach is the "SeeHear" chip from Caltech.
Other visual-auditory substitution devices deviate from the vOICe's greyscale mapping of images. Zach Capalbo's Kromophone uses a basic color spectrum correlating to different sounds and timbres to give users perceptual information beyond the vOICe's capabilities.
Nervous system implantsEdit
By means of stimulating electrodes implanted into the human nervous system, it is possible to apply current pulses to be learned and reliably recognized by the recipient. It has been shown successfully in experimentation, by Kevin Warwick, that signals can be employed from force/touch indicators on a robot hand as a means of communication.
It has been argued that the term "substitution" is misleading, as it is merely an "addition" or "supplementation" not a substitution of a sensory modality.
Building upon the research conducted on sensory substitution, investigations into the possibility of augmenting the body's sensory apparatus are now beginning. The intention is to extend the body's ability to sense aspects of the environment that are not normally perceivable by the body in its natural state.
The findings of research into sensory augmentation (as well as sensory substitution in general) that investigate the emergence of perceptual experience (qualia) from the activity of neurons have implications for the understanding of consciousness.
In 2005, the feelSpace group conducted a study of sensory augmentation with a vibrotactile magnetic compass belt worn around the waist. In this study, the participants were provided with the direction of magnetic north as a vibration on their waist.
Significant performance improvements in navigational tests were observed (over and above those experienced by control subjects during the same period with the same training) and, for half of the participants, the perception of the belt's vibration underwent a profound change from simple tactile innervation to approach a genuine and direct sense of allocentric orientation: in other words, could perceive north as an entity distinct from the vibrating transducer on the waist, like one perceives a glass on a table as an entity distinct from the impact of reflected photons on the retina. Further, tests of the influence of the belt information on the rotational nystagmus effect suggested that, after training, the processing of the belt information became subcognitive.
- Bach-y-Rita P, Collins CC, Saunders F, White B, Scadden L (1969). "Vision substitution by tactile the image projection". Nature. 221 (5184): 963–964. Bibcode:1969Natur.221..963B. doi:10.1038/221963a0. PMID 5818337.
- Nicholas Humphrey (1999). A History of the Mind: Evolution and the Birth of Consciousness. Springer. ISBN 978-0-387-98719-4.
- Scheff, Chaim-Meyer (1 January 1986). "Experimental Model for the Study of Changes in the Organization of Human Sensory Information Processing Through the Design and Testing of Non-invasive Prosthetic Devices for Sensory Impaired People". SIGCAPH Comput. Phys. Handicap. (36): 3–10. doi:10.1145/15711.15713.
- Bach-y-Rita P (2004). "Tactile sensory substitution studies". Annals of the New York Academy of Sciences. 1013 (1): 83–91. Bibcode:2004NYASA1013...83B. doi:10.1196/annals.1305.006. PMID 15194608.
- Renier L, De Volder AG (2005). "Cognitive and brain mechanisms in sensory substitution of vision: a contribution to the study of human perception". Journal of Integrative Neuroscience. 4 (4): 489–503. doi:10.1142/S0219635205000999. PMID 16385643.
- Bach-y-Rita P, Kercel SW (2003). "Sensory substitution and the human-machine interface" (PDF). Trends in Cognitive Sciences. 7 (12): 541–546. CiteSeerX 10.1.1.159.9777. doi:10.1016/j.tics.2003.10.013. PMID 14643370.
- O'Regan, JK; Noe, A. (2001). "A sensorimotor account of vision and visual consciousness". Behavioral and Brain Sciences. 24 (5): 939–973. doi:10.1017/s0140525x01000115. PMID 12239892.
- Bach-y-Rita P. Brain Mechanisms in Sensory Substitution, Academic Press New York:1972.
- Bach-y-Rita P. Nonsynaptic Diffusion Neurotransmission and Late Brain Reorganization, Demos-Vermande, New York :1995.
- Van Boven, R. W.; Hamilton, R. H.; Kauffman, T.; Keenan, J. P.; Pascual-Leone, A. (2000-06-27). "Tactile spatial resolution in blind braille readers". Neurology. 54 (12): 2230–2236. doi:10.1212/wnl.54.12.2230. ISSN 0028-3878. PMID 10881245.
- Goldreich, Daniel; Kanics, Ingrid M. (2003-04-15). "Tactile acuity is enhanced in blindness". The Journal of Neuroscience. 23 (8): 3439–3445. doi:10.1523/JNEUROSCI.23-08-03439.2003. ISSN 1529-2401. PMID 12716952.
- Goldreich, Daniel; Kanics, Ingrid M. (November 2006). "Performance of blind and sighted humans on a tactile grating detection task". Perception & Psychophysics. 68 (8): 1363–1371. doi:10.3758/bf03193735. ISSN 0031-5117. PMID 17378422.
- Wong, Michael; Gnanakumaran, Vishi; Goldreich, Daniel (2011-05-11). "Tactile Spatial Acuity Enhancement in Blindness: Evidence for Experience-Dependent Mechanisms". Journal of Neuroscience. 31 (19): 7028–7037. doi:10.1523/jneurosci.6461-10.2011. PMID 21562264.
- Bhattacharjee, Arindam; Ye, Amanda J.; Lisak, Joy A.; Vargas, Maria G.; Goldreich, Daniel (2010-10-27). "Vibrotactile Masking Experiments Reveal Accelerated Somatosensory Processing in Congenitally Blind Braille Readers". Journal of Neuroscience. 30 (43): 14288–14298. doi:10.1523/jneurosci.1447-10.2010. PMC 3449316. PMID 20980584.
- Poirier C, De Volder AG, Scheiber C (2007). "What neuroimaging tells us about sensory substitution". Neuroscience & Biobehavioral Reviews. 31 (7): 1064–1070. doi:10.1016/j.neubiorev.2007.05.010. PMID 17688948.
- Vallbo AB, Johansson RS (1984). "Properties of cutaneous mechanoreceptors in the human hand related to touch sensation". Human Neurobiology. 3 (1): 3–14. PMID 6330008.
- Kaczmarek KA, Webster JG, Bach-y-Rita P, Tompkins WJ (1991). "Electrotactile and vibrotactile displays for sensory substitution systems". IEEE Transactions on Biomedical Engineering. 38 (1): 1–16. doi:10.1109/10.68204. PMID 2026426.
- Biswas, Abhijit; Manivannan, M.; Srinivasan, Mandyam A. (2015). "Vibrotactile Sensitivity Threshold: Nonlinear Stochastic Mechanotransduction Model of the Pacinian Corpuscle". IEEE Transactions on Haptics. 8 (1): 102–113. doi:10.1109/TOH.2014.2369422. PMID 25398183.
- Biswas, Abhijit; Manivannan, M.; Srinivasan, Mandyam A. (2015). "Multiscale Layered Biomechanical Model of the Pacinian Corpuscle". IEEE Transactions on Haptics. 8 (1): 31–42. doi:10.1109/TOH.2014.2369416. PMID 25398182.
- Bach-y-Rita P, Kaczmarek KA, Tyler ME, Garcia-Lara J (1998). "Form perception with a 49-point electrotactile stimulus array on the tongue: a technical note" (PDF). J Rehabil Res Dev. 35 (4): 427–30. PMID 10220221. See also Brainport
- Layton, Julia. "How BrainPort Works". HowStuffWorks. Retrieved July 21, 2016.
- "Wicab Announces European Market Approval for its Non-Invasive Assistive Aid for the Blind" (PDF) (Press Release). Wicab, Inc. Archived from the original (PDF) on May 7, 2013.
- Hui Tang; D. J. Beebe (2003). "Design and microfabrication of a flexible oral electrotactile display". Journal of Microelectromechanical Systems. 12 (1): 29–36. doi:10.1109/JMEMS.2002.807478.
- Deyle, Travis (August 11, 2010). "Electrotactile Arrays for Texture and Pressure Feedback During Robotic Teleoperation". Hizook. Retrieved July 21, 2016.
- S. GRIMNES, Electrovibration, cutaneous sensation of microampere current, doi:10.1111/j.1748-1716.1983.tb07235.x pdf
- Kurt A. Kaczmarek; Krishnakant Nammi; Abhishek K. Agarwal; Mitchell E. Tyler; Steven J. Haase; David J. Beebec (2006). "Polarity effect in electrovibration for tactile display". IEEE Trans Biomed Eng. 53 (10): 2047–2054. doi:10.1109/TBME.2006.881804. PMC 2582732. PMID 17019869.
- Zeng; et al. (2015). "Interactive Audio-haptic Map Explorer on a Tactile Display". Interacting with Computers. 27 (4): 413–429. doi:10.1093/iwc/iwu006.
- Zeng; et al. (2012). Exploration and avoidance of surrounding obstacles for the visually impaired. Proc. of ACM ASSETS 2012. pp. 111–118. doi:10.1145/2384916.2384936.
- Bach-y-Rita P, and Kaczmarek KA. (2002). Tongue placed tactile output device. US Patent 6,430,450.
- T. McDaniel; S. Krishna; V. Balasubramanian,; D. Colbry; S. Panchanathan (2008). Using a haptic belt to convey non-verbal communication cues during social interactions to individuals who are blind. IEEE International Workshop on Haptic, Audio and Visual Environments and Games, 2008. HAVE 2008. pp. 13–18. doi:10.1109/HAVE.2008.4685291.
- Paul, Sathish Kumar; Rekha, V.; Sivarasu, Sudesh (2012). "Tactile Sensing Fabrics for Detecting Impairments in Leprosy Patients". Appropriate Healthcare Technologies for Developing Countries – AHT2012. The 7th International Conference – World Health and Wellbeing. London, UK: IEEE. doi:10.1049/cp.2012.1461.
- S. Krishna; S. Bala; T. McDaniel; S. McGuire; S. Panchanathan (2010). VibroGlove: an assistive technology aid for conveying facial expressions (PDF). Proceedings of the 28th of the international conference extended abstracts on Human factors in computing systems. Atlanta, Georgia, USA: ACM. pp. 3637–3642. doi:10.1145/1753846.1754031.
- "The Haptic Radar / Extended Skin Project". Ishikawa Watanabe Laboratory.
- A. Cassinelli; E. Sampaio; S.B. Joffily; H.R.S. Lima; B.P.G.R. Gusmão (2014). "Do blind people move more confidently with the Tactile Radar?" (PDF). IOS Press, Technology and Disability. 26 (2–3): 161–170. doi:10.3233/TAD-140414. Archived from the original (PDF) on 2017-03-14. Retrieved 2016-07-21.
- Eagleman, David (2015). Can we create new senses for humans? TED talks.
- NeoSensory, Inc
- Could This Futuristic Vest Give Us a Sixth Sense?, Smithsonian Magazine, Apr 2018.
- Meet the man who wants to give humans a sixth sense, The Telegraph, Jan 2019.
- Schurmann M, Caetano G, Hlushchuk Y, Jousmaki V, Hari R (2006). "Touch activates human auditory cortex". NeuroImage. 30 (4): 1325–1331. doi:10.1016/j.neuroimage.2005.11.020. PMID 16488157.
- "Sense organs synthesizer United States Patent Application 20020173823" (PDF).
- Tyler M, Danilov Y, Bach-y-Rita P (2003). "Closing an open-loop control system: vestibular substitution through the tongue". Journal of Integrative Neuroscience. 2 (2): 159–164. doi:10.1142/S0219635203000263. PMID 15011268.
- Bach-y-Rita P (1999). "Theoretical aspects of sensory substitution and of neurotransmission-related reorganization in spinal cord injury". Spinal Cord. 37 (7): 465–474. doi:10.1038/sj.sc.3100873. PMID 10438112.
- Riso RR (1999). "Strategies for providing upper extremity amputees with tactile and hand position feedback – moving closer to the bionic arm" (PDF). Technology and Health Care. 7 (6): 401–409. PMID 10665673. Archived from the original (PDF) on 2010-06-27. Retrieved 2016-07-21.
- Meijer PBL (1992). "An Experimental System for Auditory Image Representations". IEEE Transactions on Biomedical Engineering. 39 (2): 112–121. doi:10.1109/10.121642. PMID 1612614.
- Neural Regeneration Research, a peer-reviewed journal article, Use of sensory substitution devices as a model system for investigating cross-modal neuroplasticity in humans
- Auditory scene analysis and sonified visual images. Does consonance negatively impact on object formation when using complex sonified stimuli? Original Research ARTICLE, Frontiers in Psychology, 13 October 2015
- A Nature Scientific Report, Auditory Sensory Substitution is Intuitive and Automatic with Texture Stimuli
- "Computer Program Allows the Blind to 'See' With Sound". Science | AAAS. 2014-03-06. Retrieved 2017-04-15.
- "Device Trains Blind People To 'See' By Listening". Popular Science. Retrieved 2017-04-15.
- A. Amedi; W. Stern; J. A. Camprodon; F. Bermpohl; L. Merabet; S. Rotman; C. Hemond; P. Meijer; A. Pascual-Leone (June 2007). "Shape conveyed by visual-to-auditory sensory substitution activates the lateral occipital complex". Nature Neuroscience. 10 (6): 687–689. doi:10.1038/nn1912. PMID 17515898.
- M. Auvray; S. Hanneton; J. K. O'Regan (2007). "Learning to perceive with a visuo-auditory substitution system: Localisation and object recognition with 'The vOICe'" (PDF). Perception. 36 (3): 416–430. doi:10.1068/p5631. PMID 17455756.
- M. J. Proulx; P. Stoerig; E. Ludowig; I. Knoll (March 2008). "Seeing 'Where' through the Ears: Effects of Learning-by-Doing and Long-Term Sensory Deprivation on Localization Based on Image-to-Sound Substitution". PLoS ONE. 3 (3): e1840. Bibcode:2008PLoSO...3.1840P. doi:10.1371/journal.pone.0001840. PMC 2267489. PMID 18364998.
- Article at the American Academy of Ophthalmology website, Humayun: Restoring ‘Vision’ With Sensory Substitution Devices
- Murphy, Matthew C.; Nau, Amy C.; Fisher, Christopher; Kim, Seong-Gi; Schuman, Joel S.; Chan, Kevin C. (January 2016). "Top-down influence on the visual cortex of the blind during sensory substitution". NeuroImage. 125: 932–940. doi:10.1016/j.neuroimage.2015.11.021. PMC 5536833. PMID 26584776.
- How the brain's neuroplasticity lets us substitute one sense for another—and invent new ones Rewiring the Brain to Create New Senses
- Research on synesthesia has led to devices that blur the lines between the senses, and may offer new hope for the blind Sensory Substitution
- "The vOICe for Android - Android Apps on Google Play". play.google.com. Retrieved 2017-04-15.
- "Glasses which turn video into noise are allowing blind people to 'see'". Mail Online. Retrieved 2017-04-15.
- Doward, Jamie (2014-12-06). "vOICe: the soundscape headsets that allow blind people to 'see' the world". The Guardian. ISSN 0261-3077. Retrieved 2017-04-15.
- Chaturvedi, Pooja (22 Jan 2013). "Shooting blind". Livemint.com. Retrieved 16 April 2017.
- Relevant portion evidencing how far the technology has travelled and empowered the differently-abled, "... Delhi-based photographer Pranav Lal, who is also blind, uses vOICe, a camera-based visual sound technology ... experience of live camera views ... image-to-sound rendering ..." .
- Relevant portion evidencing how far the technology has travelled and empowered the differently-abled, "... Delhi-based photographer Pranav Lal, who is also blind, uses vOICe, a camera-based visual sound technology ... experience of live camera views ... image-to-sound rendering ..." .
- Gurung, Regina (15 Nov 2016). "He hears shapes, clicks it!". The New Indian Express. Retrieved 16 April 2017.
- "The vOICe - Seeing with Sound". The vOICe - Seeing with Sound. Retrieved 2017-04-15.
- "Meet the Woman Who Can See With Her Ears". Wired.com. Retrieved 8 January 2018.
- "Computer Program Allows the Blind to 'See' With Sound". Sciencemag.org. 6 March 2014. Retrieved 8 January 2018.
- Doward, Jamie (7 December 2014). "vOICe: the soundscape headsets that allow blind people to 'see' the world". Theguardian.com. Retrieved 8 January 2018.
- "Device Trains Blind People To 'See' By Listening". Popsci.com. Retrieved 8 January 2018.
- "Article Detail - International Journal of Advanced Research". International Journal of Advanced Research. doi:10.21474/ijar01/2304#sthash.usfrrruw.dpuf (inactive 2019-03-14).
- Debargha Ganguly (2016). "PROJECT BAT-EYE -Developing an Economic System that can give a Blind Person Basic Spatial Awareness and Object Identification (Updated ) (PDF Download Available)". ResearchGate. doi:10.13140/rg.2.2.12296.62729.
- Debargha Ganguly (2018). "PROJECT " AWAAZ " - Developing an extremely low cost, non-movement restrictive plugin for hand movement to speech conversion (PDF Download Available)". ResearchGate. doi:10.13140/rg.2.2.19059.94243.
- "Project Basics- a concerned effort to improve the basic standard of living of people of developing and underdeveloped countries. by Debargha Ganguly - Research Project on ResearchGate". ResearchGate. Retrieved 2017-02-16.
- Levy-Tzedek, Shelly; Hanassy S.; Abboud S.; Maidenbaum S.; Amedi A. (January 1, 2012). "Fast, accurate reaching movements with a visual-to-auditory sensory substitution device" (PDF). Restorative Neurology and Neuroscience. 30 (4): 313–323. doi:10.3233/RNN-2012-110219. PMID 22596353.
- Abboud, Sami; Hanassy S; Levy-Tzedek S; Maidenbaum S; Amedi A. (2014). "EyeMusic: Introducing a "visual" colorful experience for the blind using auditory sensory substitution" (PDF). Restorative Neurology and Neuroscience. 32 (2): 247–257. doi:10.3233/RNN-130338. PMID 24398719.
- Maidenbaum, Shachar; Abboud S.; Amedi A. (April 2014). "Sensory substitution: Closing the gap between basic research and widespread practical visual rehabilitation" (PDF). Neuroscience & Biobehavioral Reviews. 41: 3–15. doi:10.1016/j.neubiorev.2013.11.007. PMID 24275274.
- Ambard, Maxime; Benezeth Y.; Pfister P. (2015). "Mobile video-to-audio transducer and motion detection for sensory substitution". Frontiers in ICT. 2. doi:10.3389/fict.2015.00020.
- Capelle C, Trullemans C, Arno P, Veraart C (1998). "A real-time experimental prototype for enhancement of Vision Rehabilitation–vision rehabilitation using auditory substitution". IEEE Transactions on Biomedical Engineering. 45 (10): 1279–1293. doi:10.1109/10.720206. PMID 9775542.
- Nielson L, Mahowald M, Mead C (1989). "SeeHear," in Analog VLSI and Neural Systems, by C. Mead, Reading: Addison-Wesley, chapter 13, 207–227.
- [dead link]
- Warwick K, Gasson M, Hutt B, Goodhew I, Kyberd P, Schulzrinne H, Wu X (2004). "Thought communication and control: A first step using radiotelegraphy". IEEE Proceedings on Communications. 151 (3): 185–189. doi:10.1049/ip-com:20040409.
- Lenay C, Gapenne O, Hanneton S, Marque C, Geouelle C (2003). "Sensory Substitution: limits and perspectives". Touching for Knowing, Cognitive psychology of haptic manual perception (PDF). pp. 275–292.
- "Archived copy". Archived from the original on 2014-08-10. Retrieved 2014-08-06.CS1 maint: Archived copy as title (link)
- Nagel SK, Carl C, Kringe T, Märtin R, König P (2005). "Beyond sensory substitution – learning the sixth sense". Journal of Neural Engineering. 2 (4): R13–26. doi:10.1088/1741-2560/2/4/r02. PMID 16317228.
- Tongue display for sensory substitution
- The vOICe auditory display for sensory substitution.
- Artificial Retinas
- Sensory Substitution:limits and perspectives C. Lenay et al.
- The Vibe
- feelSpace - The Magnetic Perception Group of the University of Osnabrück
- The Kromophone
- Sensory Substitution For Blind (Nihat Erim İnceoğlu)