Radiocarbon dating methods produce data that must then be further manipulated in order to calculate a resulting "radiocarbon age".

## Calculations

The calculations to be performed on the measurements taken depend on the technology used, since beta counters measure the sample's radioactivity, whereas accelerator mass spectrometers (AMS) determine the ratio of the three different carbon isotopes in the sample.[1]

### Standards

The calculations to convert measured data to an estimate of the age of the sample require the use of several standards. One of these, the standard for normalizing δ13C values, is Pee Dee Belemnite (PDB), a fossil which has a 13
C
/12
C
ratio of 1.12372%.[2] A related standard is the use of wood, which has a δ13C of -25‰, as the material for which radiocarbon ages are calibrated. Since different materials have different δ13C values, it is possible for two samples of different materials, of the same age, to have different levels of radioactivity and different 14
C
/12
C
ratios. To compensate for this, the measurements are converted to the activity, or isotope ratio, that would have been measured if the sample had been made of wood. This is possible because the δ13C of wood is known, and the δ13C of the sample material can be measured, or taken from a table of typical values. The details of the calculations for beta counting and AMS are given below.[3]

Another standard is the use of 1950 as "present", in the sense that a calculation that shows that a sample's likely age is 500 years "before present" means that it is likely to have come from about the year 1450. This convention is necessary in order to keep published radiocarbon results comparable to each other; without this convention, a given radiocarbon result would be of no use unless the year it was measured was also known—an age of 500 years published in 2010 would indicate a likely sample date of 1510, for example. In order to allow measurements to be converted to the 1950 baseline, a standard activity level is defined for the radioactivity of wood in 1950. Because of the fossil fuel effect, this is not actually the activity level of wood from 1950; the activity would have been somewhat lower.[4] The fossil fuel effect was eliminated from the standard value by measuring wood from 1890, and using the radioactive decay equations to determine what the activity would have been at the year of growth. The resulting standard value, Aabs, is 226 becquerels per kilogram of carbon.[5]

Both beta counting and AMS measure standard samples as part of their methodology. These samples contain carbon of a known activity.[6] The first standard, Oxalic Acid SRM 4990B, also referred to as HOxI, was a 1,000 lb batch of oxalic acid created in 1955 by the National Institute of Standards and Technology (NIST). Since it was created after the start of atomic testing, it incorporates bomb carbon, so measured activity is higher than the desired standard. This is addressed by defining the standard to be 0.95 times the activity of HOxI.[5]

All of this first standard has long since been consumed, and later standards have been created, each of which has a given ratio to the desired standard activity. A secondary standard, Oxalic Acid SRM 4990C, also referred to as HOxII, 1,000 lb of which was prepared by NIST in 1977 from French beet harvests, is now in wide use.[7]

### Calculations for beta counting devices

To determine the age of a sample whose activity has been measured by beta counting, the ratio of its activity to the activity of the standard must be found. The equation:[8]

${\displaystyle A_{s}=A_{std}\left({\frac {M_{s}-M_{b}}{M_{std}-M_{b}}}\right)}$

gives the required ratio, where As is the true activity of the sample, Astd is the true activity of the standard, Ms is the measured activity of the sample, Mstd is the measured activity of the standard, and Mb is the measured activity of the blank.[8]

A correction must also be made for fractionation. The fractionation correction converts the 14
C
/12
C
ratio for the sample to the ratio it would have had if the material was wood, which has a δ13C value of -25‰. This is necessary because determining the age of the sample requires a comparison of the amount of 14
C
in the sample with what it would have had if it newly formed from the biosphere. The standard used for modern carbon is wood, with a baseline date of 1950.[3]

Correcting for fractionation changes the activity measured in the sample to the activity it would have if it were wood of the same age as the sample. The calculation requires the definition of a 13
C
fractionation factor, which is defined for any sample material as[4]

${\displaystyle {\ce {Frac}}_{{\ce {13/12 (sample)}}}={\frac {\left({\frac {{\ce {^{13}C}}}{{\ce {^{12}C}}}}\right)_{{\ce {wood}}}}{\left({\frac {{\ce {^{13}C}}}{{\ce {^{12}C}}}}\right)_{{\ce {sample}}}}}}$

The 14
C
fractionation factor, Frac14/12, is approximately the square of this, to an accuracy of 1‰:[4]

${\displaystyle {\text{Frac}}_{\text{14/12 (sample)}}=({\text{Frac}}_{\text{13/12 (sample)}})^{2}}$

Multiplying the measured activity for the sample by the 14
C
fractionation factor converts it to the activity that it would have had had the sample been wood:[4]

${\displaystyle A_{sn}=A_{s}{\text{Frac}}_{14/12(s)}}$

where Asn is the normalized activity for the sample, and Frac14/12 (s) is the 14
C
fractionation factor for the sample.[4]

The equation for δ13C given earlier can be rearranged to[4]

${\displaystyle \left({\frac {{\ce {^{13}C}}}{{\ce {^{12}C}}}}\right)_{{\ce {sample}}}=\left(1+{\frac {{\ce {\delta^{13}C}}}{1000}}\right)\left({\frac {{\ce {^{13}C}}}{{\ce {^{12}C}}}}\right)_{{\ce {PDB}}}}$

Substituting this in the 14
C
fractionation factor, and also substituting the value for δ13C for wood of -25‰, gives the following expression:[4]

${\displaystyle A_{sn}=A_{s}\left({\frac {\left(1-{\frac {25}{1000}}\right)\left({\frac {{\ce {^{13}C}}}{{\ce {^{12}C}}}}\right)_{{\ce {PDB}}}}{\left(1+{\frac {{\ce {\delta^{13}C}}}{1000}}\right)\left({\frac {{\ce {^{13}C}}}{{\ce {^{12}C}}}}\right)_{{\ce {PDB}}}}}\right)^{2}}$

where the δ13C value remaining in the equation is the value for the sample itself. This can be measured directly, or simply looked up in a table of characteristic values for the type of sample material—this latter approach leads to increased uncertainty in the result, as there is a range of possible δ13C values for each possible sample material. Cancelling the PDB 13
C
/12
C
ratio reduces this to:[4]

${\displaystyle A_{sn}=A_{s}\left({\frac {\left(1-{\frac {25}{1000}}\right)}{\left(1+{\frac {{\ce {\delta^{13}C}}}{1000}}\right)}}\right)^{2}}$

### AMS calculations

The results from AMS testing are in the form of ratios of 12
C
, 13
C
, and 14
C
. These ratios are used to calculate Fm, the "fraction modern", defined as

${\displaystyle F_{m}={\frac {R_{norm}}{R_{modern}}}}$

where Rnorm is the 14
C
/12
C
ratio for the sample, after correcting for fractionation, and Rmodern is the standard 14
C
/12
C
ratio for modern carbon.[9]

The calculation begins by subtracting the ratio measured for the machine blank from the other sample measurements. That is:

${\displaystyle R'_{s}=R_{s}-R_{mb}}$
${\displaystyle R'_{std}=R_{std}-R_{mb}}$
${\displaystyle R'_{pb}=R_{pb}-R_{mb}}$

where Rs is the measured sample 14
C
/12
C
ratio; Rstd is the measured ratio for the standard; Rpb is the measured ratio for the process blank, and Rmb is the measured ratio for the machine blank. The next step, to correct for fractionation, can be done using either the 14
C
/12
C
ratio or the 14
C
/13
C
ratio, and also depends on which of the two possible standards was measured: HOxI or HoxII. R'std is then R'HOxI or R'HOxII, depending on which standard was used. The four possible equations are as follows. First, if the 14
C
/12
C
ratio is used to perform the fractionation correction, the following two equations apply, one for each standard.[9]

${\displaystyle R_{{\ce {HOxI,-19}}}=R'_{{\ce {HoxI}}}\left({\frac {1+{\frac {-19}{1000}}}{1+{\frac {{\ce {\delta^{13}C_{HoXI}}}}{1000}}}}\right)^{2}}$
${\displaystyle R_{{\ce {HOxII,-25}}}=R'_{{\ce {HoxII}}}\left({\frac {1+{\frac {-25}{1000}}}{1+{\frac {{\ce {\delta^{13}C_{HoXII}}}}{1000}}}}\right)^{2}}$

If the 14
C
/13
C
ratio is used instead, then the equations for each standard are:[9]

${\displaystyle R_{{\ce {HOxI,-19}}}=R'_{{\ce {HoxI}}}\left({\frac {1+{\frac {-19}{1000}}}{1+{\frac {{\ce {\delta^{13}C_{HoXI}}}}{1000}}}}\right)}$
${\displaystyle R_{{\ce {HOxII,-25}}}=R'_{{\ce {HoxII}}}\left({\frac {1+{\frac {-25}{1000}}}{1+{\frac {{\ce {\delta^{13}C_{HoXII}}}}{1000}}}}\right)}$

The δ13C values in the equations measure the fractionation in the standards as CO
2
, prior to their conversion to graphite to use as a target in the spectrometer. This assumes that the conversion to graphite does not introduce significant additional fractionation.[9]

Once the appropriate value above has been calculated, Rmodern can be determined; it is[9]

${\displaystyle R_{modern}=0.95R_{\text{HOxI,-19}}=.7459R_{\text{HOxII,-25}}}$

The values 0.95 and 0.7459 are part of the definition of the two standards; they convert the 14
C
/12
C
ratio in the standards to the ratio that modern carbon would have had in 1950 if there had been no fossil fuel effect.[9]

Since it is common practice to measure the standards repeatedly during an AMS run, alternating the standard target with the sample being measured, there are multiple measurements available for the standard, and these measurements provide a couple of options in the calculation of Rmodern. Different labs use this data in different ways; some simply average the values, while others consider the measurements made on the standard target as a series, and interpolate the readings that would have been measured during the sample run, if the standard had been measured at that time instead.[9]

Next, the uncorrected fraction modern is calculated; "uncorrected" means that this intermediate value does not include the fractionation correction.[9]

${\displaystyle Fm_{uc}={\frac {R'_{s}}{R_{modern}}}}$

Now the measured fraction modern can be determined, by correcting for fractionation. As above there are two equations, depending on whether the 14
C
/12
C
or 14
C
/13
C
ratio is being used. If the 14
C
/12
C
ratio is being used:[9]

${\displaystyle Fm_{ms}=Fm_{uc}\left({\frac {1+{\frac {-25}{1000}}}{1+{\frac {{\ce {\delta^{13}C}}_{s}}{1000}}}}\right)^{2}}$

If the 14
C
/13
C
ratio is being used:[9]

${\displaystyle Fm_{ms}=Fm_{uc}\left({\frac {1+{\frac {-25}{1000}}}{1+{\frac {{\ce {\delta^{13}C}}_{s}}{1000}}}}\right)}$

The δ13Cs value is from the sample itself, measured on CO
2
prepared while converting the sample to graphite.[9]

The final step is to adjust Fmms for the measured fraction modern of the process blank, Fmpb, which is calculated as above for the sample. One approach[note 1] is to determine the mass of the measured carbon, Cms, along with Cpb, the mass of the process blank, and Cs, the mass of the sample. The final fraction modern, Fms is then[9]

${\displaystyle Fm_{s}={\frac {Fm_{ms}{\ce {C}}_{ms}-Fm_{pb}{\ce {C}}_{pb}}{{\ce {C}}_{s}}}}$

The fraction modern is then converted to an age in "radiocarbon years", meaning that the calculation uses Libby's half-life of 5,568 years, not the more accurate modern value of 5,730 years, and that no calibration has been done:[10]

${\displaystyle {\text{Age}}=-8033\ln(Fm)}$

### Errors and reliability

There are several possible sources of error in both the beta counting and AMS methods, although laboratories vary in how they report errors. All laboratories report counting statistics—that is, statistics showing possible errors in counting the decay events or number of atoms—with an error term of 1σ (i.e. 68% confidence that the true value is within the given range).[11] These errors can be reduced by extending the counting duration: for example, testing a modern benzene sample will find about eight decay events per minute per gram of benzene, and 250 minutes of counting will suffice to give an error of ± 80 years, with 68% confidence. If the benzene sample contains carbon that is about 5,730 years old (the half-life of 14
C
), then there will only be half as many decay events per minute, but the same error term of 80 years could be obtained by doubling the counting time to 500 minutes.[12][13] Note that the error term is not symmetric, though the effect is negligible for recent samples; for a sample with an estimated age of 30,600 years, the error term might be +1600 to -1300.[11]

To be completely accurate, the error term quoted for the reported radiocarbon age should incorporate counting errors not only from the sample, but also from counting decay events for the reference sample, and for blanks. It should also incorporate errors on every measurement taken as part of the dating method, including, for example, the δ13C term for the sample, or any laboratory conditions being corrected for such as temperature or voltage. These errors should then be mathematically combined to give an overall term for the error in the reported age, but in practice laboratories differ, not only in the terms they choose to include in their error calculations, but also in the way they combine errors. The resulting 1σ estimates have been shown to typically underestimate the true error, and it has even been suggested that doubling the given 1σ error term results in a more accurate value.[11][12]

The usual presentation of a radiocarbon date, as a specific date plus or minus an error term, obscures the fact that the true age of the object being measured may lie outside the range of dates quoted. In 1970, the British Museum radiocarbon laboratory ran weekly measurements on the same sample for six months. The results varied widely (though consistently with a normal distribution of errors in the measurements), and included multiple date ranges (of 1σ confidence) that did not overlap with each other. The extreme measurements included one with a maximum age of under 4,400 years, and another with a minimum age of over 4,500 years.[14]

It is also possible for laboratories to have systematic errors, caused by weaknesses in their methodologies. For example, if 1% of the benzene in a modern reference sample is allowed to evaporate, scintillation counting will give a radiocarbon age that is too young by about 80 years. Laboratories work to detect these errors both by testing their own procedures, and by periodic inter-laboratory comparisons of a variety of different samples; any laboratories whose results differ from the consensus radiocarbon age by too great an amount may be suffering from systematic errors. Even if the systematic errors are not corrected, the laboratory can estimate the magnitude of the effect and include this in the published error estimates for their results.[15]

The limit of measurability is approximately eight half-lives, or about 45,000 years. Samples older than this will typically be reported as having an infinite age. Some techniques have been developed to extend the range of dating further into the past, including isotopic enrichment, or large samples and very high precision counters. These methods have in some cases increased the maximum age that can be reported for a sample to 60,000 and even 75,000 years.[16][17]

## Notes

1. ^ McNichol & Burr give two other calculations, one of which can be shown to be equivalent to the one given here. The other depends on the process blank being the same mass as the sample.[9]

## Footnotes

1. ^ McNichol, Jull & Burr, "Converting AMS Data to Radiocarbon Values: Considerations and Conventions", pp. 313.
2. ^ Dass (2007), p.276.
3. ^ a b Aitken, Science-based Dating in Archaeology, p. 92–95.
4. Eriksson Stenstrom et al. A Guide to Radiocarbon Units and Calculations, p. 6.
5. ^ a b L'Annunziata, Radioactivity, p. 528.
6. ^ Aitken, Science-based Dating in Archaeology, pp. 82-85.
7. ^ J. Terasmae, "Radiocarbon Dating: Some Problems and Potential Developments", in Mahaney, Quaternary Dating Methods, p. 5.
8. ^ a b Eriksson Stenström et al., "A guide to radiocarbon units and calculations", p. 3.
9. McNichol, Jull & Burr, "Converting AMS Data to Radiocarbon Values: Considerations and Conventions", pp. 315-318.
10. ^ "Radiocarbon Data Calculations: NOSAMS". Woods Hole Oceanographic Institution. 2007. Retrieved August 27, 2013.
11. ^ a b c Taylor, Radiocarbon Dating, p. 102−104.
12. ^ a b Bowman, Radiocarbon Dating, pp. 38–39.
13. ^ Taylor, Radiocarbon Dating, p. 124.
14. ^ Taylor, Radiocarbon Dating, pp. 125−126.
15. ^ Bowman, Radiocarbon Dating, pp. 40−41.
16. ^ Mike Walker, Department of Archaeology and Anthropology, University of Wales, Lampeter, UK Quaternary Dating Methods, John Wiley & Sons, Ltd. 2005, page 23. Online, pdf 9 MByte Archived 2014-07-14 at the Wayback Machine
17. ^ Shennan, Ian; Long, Antony; Horton, Benjamin (2014). Handbook of Sea-Level Research. John Wiley & Sons. p. 350. ISBN 1118452569. Retrieved 31 October 2018.