Radiation Portal Monitor
||The examples and perspective in this article may not represent a worldwide view of the subject. (December 2010)|
Radiation Portal Monitors (RPMs) are passive radiation detection devices used for the screening of individuals, vehicles, cargo or other vectors for detection of illicit sources such as at borders or secure facilities. Fear of terrorist attacks with radiological weapons spurred RPM deployment for cargo scanning since 9/11, particularly in the United States.
RPMs were originally developed for screening individuals and vehicles at secure facilities such as weapons laboratories. They were deployed at scrap metal facilities to detect radiation sources mixed among scrap that could contaminate a facility and result in a costly clean up. As part of the effort to thwart nuclear smuggling after the breakup of the Soviet Union, RPMs were deployed around that territory, and later around many other European and Asian countries, by the US Department of Energy (DOE) National Nuclear Security Administration (NNSA) Second Line of Defense Program (SLD) starting in the late 1990s. After the attack of 9/11, the US Customs and Border Protection (CBP) started the Radiation Portal Monitor Program (RPMP) to deploy RPMs around all US borders (land, sea and air).
Radiation Portal Monitor (RPM) was designed to detect traces of radiation emitted from an object passing through a RPM. Gamma radiation is detected, and in some cases complemented by neutron detection when sensitivity for nuclear material is desired.
PVT (gamma ray detection)
First generation RPMs often rely on PVT scintillators for gamma counting. They provide limited information on energy of detected photons, and as a result, they were criticized for their inability to distinguish gamma rays originating from nuclear sources from gamma rays originating from a large variety of benign cargo types that naturally emit radioactivity, including cat litter, granite, porcelain, stoneware, banana etc. Those Naturally Occurring Radioactive Materials, called NORMs account for 99% of nuisance alarms. It is worth noting that bananas have erroneously been reported as the source of radiation alarms; they are not. Most produce contains potassium-40, but packing density of fruits and vegetables is too low to produce a significant signal. PVT does have the ability to provide some energy discrimination, which can be exploited to limit nuisance alarms from NORM.
NaI(Tl) (gamma ray detection)
In attempt to reduce the high nuisance alarm rates of first generation RPMs, the Advanced Spectroscopic Portal (ASP) program was called into life. Some of the portal monitors evaluated for this purposes are based on NaI(Tl) scintillating crystals. These devices, having better energy resolution than PVT, were supposed to reduce nuisance alarm rates by distinguishing threats from benign sources on the basis of the detected gamma radiation spectra. ASPs based on NaI(Tl) had a cost several times that of first generation RPMs. To date, NaI(Tl) based ASPs have not been able to demonstrate significantly better performance than PVT based RPMs.
HPGe (gamma ray detection)
In the scope of the ASP program, high purity germanium (HPGe) based portal monitors were evaluated. HPGe, having significantly better energy resolution than NaI(Tl), allows rather precise measurement of the isotopes contributing to gamma ray spectra. However, due to very high costs and major constraints such as cryo-cooling requirements, US government support for HPGe based portal monitors was dropped.
3He (thermal neutron detection)
RPMs geared for interception of nuclear threats usually incorporate a neutron detection technology. The vast majority of all neutron detectors deployed in RPMs to date relies on He-3 tubes surrounded by neutron moderators. Since the end of 2009, the global He-3 supply crisis  however has made this technology unavailable. The search for alternative neutron detection technologies has yielded satisfactory results.
RPMs are deployed with the aim to intercept radiological threats as well as to deter malicious groups from deploying such threats.
Radiological dispersal devices
Radiological dispersal devices (RDDs) are weapons of mass disruption rather than weapons of mass destruction. "Dirty bombs" are examples of RDDs. As the name suggests, an RDD aims at dispersing radioactive material over an area, causing high cleanup costs, psychological, and economic damage. Nevertheless, direct human losses caused by RDDs are low and not attributed to the radiological aspect. RDDs are easily fabricated and components readily obtainable. RDDs can are comparatively easy to detect with RPMs due to their high level of radioactivity. RDDs emit gamma radiation as well as sometimes, depending on what isotopes are used, neutrons.
Improvised nuclear devices (INDs) and nuclear weapons are weapons of mass destruction. They are difficult to acquire, manufacture, refurbish, and handle. While INDs can be constructed to emit only low amounts of radiation making them difficult to detect with RPMs, all INDs emit some amounts of gamma and neutron radiation.
Gamma radiation as well as neutron radiation can cause RPMs to trigger an alarm procedure. Alarms caused by statistical fluctuations of detection rates are referred to as false alarms. Alarms caused by benign radioactive sources are referred to as nuisance alarms. Causes of nuisance alarms can be broken up into several large categories:
- Ceramic, tiles, porcelain, pottery, granite, clay, and other rock and clay based product contain elevated levels of naturally occurring Potassium-40 and in smaller degree Thorium-232 and other isotopes
- Propane gas tankers, full or empty, contain elevated levels of Radium-226
- Many fertilizers and potash contain elevated levels of Potassium-40
- Cat litter contains elevated levels of Thorium-232
This article relates primarily to RPMs deployed for screening trucks at ports of entry. Over 1400 RPMs are deployed at US borders and a similar number at foreign locations for the purpose of interdicting illicit radiological and nuclear material. The US deployments cover all land border vehicles, all seaport containerized cargo, and all mail and express courier facilities. Efforts are also being made to deploy similar measures to other cross border vectors including:
- Pedestrian radiation portal monitoring
- Air freight radiation portal monitoring
- Crane based radiation portal monitoring
- Air luggage radiation portal monitoring
- Railway radiation portal monitoring
RPMs are also deployed at civilian and military nuclear facilities to prevent theft of radiological materials. Steel mills often use RPMs to screen incoming scrap metal to avoid radioactive sources illegally disposed in this way. Garbage incineration plants often monitor incoming material to avoid contamination.
- Fehlau, P.E., and G.S Brunson. 1983. Coping with plastic scintillators in nuclear safeguards. IEEE Transactions on Nuclear Science 30:158–161
- Second Line of Defense Program
- Kouzes, R.T., "Detecting Illicit Nuclear Materials", American Scientist 93, PP. 422-427 (September-October 2005).
- Kouzes, RT, JH Ely, PE Keller, RJ McConn, ER Siciliano. “Passive Neutron Detection for Interdiction of Nuclear Material at Borders,” NIM-A 584, 383-400 (2008)
- Waste, Abuse, and Mismanagement in Department of Homeland Security Contracts. United States House of Representatives. July 2006. pp. 12–13.
- "Manual for Ludlum Model 3500-1000 Radiation Detector System". Retrieved Sep. 2007.
- Ely, J.H., R.T. Kouzes, J. Schweppe, E.R. Siciliano, D. Strachan, D. Weier, The Use Of Energy Windowing To Discriminate SNM From NORM, NIM-A 560, Issue 2, 10 (May 2006), pp 373-387
- "Evaluating testing, costs, and benefits of advanced spectroscopic portals for screening cargo at ports of entry: interim report" (2009)
- New York Times article
- Kouzes, R.T., J.H. Ely, L.E. Erikson, W.J. Kernan, A.T. Lintereur, E.R. Siciliano, D.L. Stephens, D.C. Stromswold, R.M. VanGinhoven, M.L. Woodring, Neutron Detection Alternatives For Homeland Security, Nuclear Instruments and Methods in Physics Research A 623 (2010) 1035–1045
- Kouzes, RT, JH Ely, J Evans, W Hensley, E Lepel, J McDonald, J Schweppe, ER Siciliano, D Strom, M Woodring, “Naturally Occurring Radioactive Materials In Cargo At U.S. Borders,” Packaging, Transport, Storage & Security of Radioactive Material, vol. 17, no. 1 pp 11-17, 2006
- Domestic Nuclear Detection Office, " Radiation Quick Reference Guide" 
- Cooley, Geri. "NORM Management in the Oilfield". Permian Basin STEPS Network October Industry Meeting, October 14, 2008. 
- Kouzes RT and ER Siciliano. “The Response of Radiation Portal Monitors to Medical Radionuclides,” in Radiation Measurements 41, pp 499-512 (May 2006)